APC-7 connector

From Wikipedia, the free encyclopedia

Jump to: navigation, search

The APC-7 connector (for Amphenol Precision Connector) is a precision sexless, 7 mm, coaxial connector that was used on laboratory microwave test equipment up to 18 GHz. The connector is not suitable for modern laboratory equipment that works above 18 GHz, so the connectors are not as common as they were.

The connectors are expensive, so they are seldom used outside the laboratory.

APC-7 connectors are used on microwave test equipment because low-reflection coefficient and repeatable coaxial connections are required. Furthermore, cost is not a central concern for test equipment: accuracy is.

An APC-7 connector pair has several desirable features.

The connectors are sexless. Sexless connectors avoid some awkward adapters or duplication of laboratory standards. With a sexed connector, a laboratory might need a precision male load and a precision female load. With a sexless connector, only one load is required.

The connectors have a well defined reference plane. Consequently, the connectors simplify some measurements.

The mated pair of APC-7 connectors has a low reflection coefficient / low VSWR. The low reflection coefficient means better measurement accuracy. The maximum VSWR is 1.04:1 at 18 GHz.

The mated pair makes a repeatable connection. A pair has a repeatable reflection coefficent to ±0.001. Consequently, laboratory measurements will be repeatable.

Contents

1 Adapters
2 Origin of the name

2.1 Other names

3 Notes
4 External links

[edit] Adapters

The features of the APC-7 connector come at a price — adapters are required to change from the laboratory world to the connectors that are used on everyday products. These adapters are precision devices and are expensive.

Adapters between APC-7 and other common RF connectors (e.g. N, SMA, TNC) are available from various vendors.

[edit] Origin of the name

APC-7 stands for Amphenol Precision Connector – 7 mm. Hewlett-Packard started developing the connector in the mid-1960s. Amphenol improved the design and started manufacturing the connector.[1]

[edit] Other names

The connector is sometimes referred to a just a 7 mm connector. (Amphenol also developed the smaller APC-3.5 connector that is an improvement on the SMA connector.)

Companies will commonly change the first letter of the acronym to match theirs. So you sometimes see APC-7 from Maury Microwave or Agilent Technologies or RPC-7 from Rosenberger Micro-Coax.

For this reason the acronym GPC-7 was introduced. This is short for General Precision Connector-7.

LPC-7 (Laboratory Precision Connector-7) refers to inner conductors with only air dielectrics. This means that LPC-7 can only refer to unsupported air lines.

As GPC-7 started to be used for lower grades of the same type, a catch all of PC-7 (Precision Connector-7) is now often used.

[edit] Notes

^ Hewlett-Packard (1979–80). Coaxial & Waveguide Catalog and Microwave Measurement Handbook. pp. 90.

[edit] External links

Rosenberger RPC-7

v • d • e

RF connectors

N · BNC · TNC · F · SMA · SMB · SMC · TV Aerial Plug · MMCX · MCX · FME · C · UHF · GR · Motorola · Musa · Mini-UHF · QMA/QN · APC-7 · EIA · Hirose U.FL

Retrieved from "http://en.wikipedia.org/wiki/APC-7_connector"

Category: Coaxial connectors

Views

Article
Discussion
Edit this page
History

Personal tools

Log in / create account

Navigation

Main page
Contents
Featured content
Current events
Random article

Search

Interaction

About Wikipedia
Community portal
Recent changes
Contact Wikipedia
Donate to Wikipedia
Help

Toolbox

What links here
Related changes
Upload file
Special pages
Printable version
Permanent link
Cite this page

This page was last modified on 5 January 2009, at 23:37.
All text is available under the terms of the GNU Free Documentation License. (See Copyrights for details.)
Wikipedia® is a registered trademark of the Wikimedia Foundation, Inc., a U.S. registered 501(c)(3) tax-deductible nonprofit charity.

Privacy policy
About Wikipedia
Disclaimers